|PRODUCT DIMENSIONS||5.1 x 3.3 x 2.2 in. (130 x 85 x 55 mm)|
|WARRANTY||One year limited warranty against defects in workmanship and materials. Warranty does not cover batteries.|
|MANUFACTURER ITEM CODE||6700|
|METHOD OF NEBULIZATION||Ultrasonic Technology|
|PRODUCT WEIGHT||.44 Lb (w/o battery)|
|PARTICLE SIZE||0.5~4.0 Microns|
Portable Ultrasonic Nebulizer is designed to be used by an adult or pediatric patient to produce medicated aerosols for inhalation therapy. It features a two-valve system that ensures maximum efficiency of inhalation, thus reducing the waste of medication. In practice, the mouthpiece expiration valve will open during expiration, allowing the patient's exhaled air to pass to the atmosphere, avoiding contamination of the device. During inspiration, the valve will close, maximizing the inhalation amount. As with all ultrasonic nebulizers, this nebulizer is not for use with Pulmicort.
You have the option to purchase this ultrasonic nebulizer with or without a rechargeable battery. With the rechargeable battery you'll have the opportunity to take treatments nearly anywhere!
It's important to regularly replace this ultrasonic nebulizer's air filter every 30 days or when it appears dirty or gray. Not regularly replacing the air filter will cause your nebulizer to under perform, and cause treatments to be less effective. In addition, this unit requires distilled water: Fill the reservoir of the device with distilled water until it reaches exactly the level marked in red (approximately 4 ml). This amount of water serves as a liquid conductor, to conduct the ultrasound waves to the medication, and will never be nebulized.
To prevent possible risk of infection from contaminated medication, clean the nebulizer as follows after each treatment.
To clean: Ensure the plug is disconnected from the device's power socket. Remove the top cover, nebulization chamber, airflow chamber, and medication from the devise. Remove the filter from the cover. Completely empty the conduction liquid from the reservoir. Wipe the main unit with a clean, soft, moistened cloth. Wash the top cover, nebulization chamber, air-flow chamber, mouthpiece, masks, and medication cup in warm, soapy water, and rinse thoroughly. Allow to air dry. Store in a clean, dry location.
Once daily, after washing as above, soak the washed components in a fresh solution consisting of one part warm water and one part white vinegar for 30 minutes followed by a warm water rinse. Allow to air dry. Store in a clean, dry location.
Bought it followed directions, as you have to add distilled water to the red mark, then add your medication, and turned it on, did not produce any mist at all, contacted Them and sent it back, but they refuse to accept it, since it was used. Even though it really wasn’t used. It’s never worked! They are supposed to be returning it to me, as they said it worked for them, but it hasn’t been returned, they where supposed to send me an email showing that it works but haven’t received that either! Don’t buy it!!!
We apologize for your experience. Our technicians received your nebulizer and found the unit to be in good operating condition. Please call technical support at 770-368-4700 for any further assistance.
Good luck trying to use it on battery power. Not gonna get long winded. Don’t waste your money
Lumiscope Portable Ultrasonic Nebulizer
I needed a nebulizer that I could take on the road and not have to lug the big bulky compressor ones. This is exactly what I was looking for!
Use it in my car. Compact