Recently added item(s)
You have no items in your shopping cart.
Most orders ship within 1 business day. Shipping times for:
Even big or heavy items, nebulizers and oxygen supplies, ship free. If your order
total is $75+, the shipping's on us!
Omron Micro-Air Electronic Nebulizer System NE-U22V1
PARI LC Sprint Reusable Nebulizer Set
Medquip Penguin Nebulizer System
Handi-Air Tote Wheeled Oxygen Carrier
Respironics EasyLife Nasal CPAP Mask
Health-Ox Fingertip Pulse Oximeter
* Required Fields
Portable Ultrasonic Nebulizer is designed to be used by an adult or pediatric patient to produce medicated aerosols for inhalation therapy. It features a two-valve system that ensures maximum efficiency of inhalation, thus reducing the waste of medication. In practice, the mouthpiece expiration valve will open during expiration, allowing the patients exhaled air to pass to atmosphere, avoiding contamination of the device. During inspiration, the valve will close, maximizing the inhalation amount. As with all ultrasonic nebulizer, this nebulizer is not for use with Pulmicort.
You have the option to purchase this ultrasonic nebulizer with or without a rechargeable battery. With the rechargeable battery you'll have the opportunity to take treatments nearly anywhere!
It's important to regularly replace this ultrasonic nebulizer's air filter every 30 days or when it appears dirty or gray. Not regularly replacing the air filter will cause you nebulizer to under perform, and cause treatments to be less effective.
You can view this nebulizer's air filters, as well as every part/accessory for the Lumiscope here.
What's in the Box: ultrasonic nebulizer, adult mask, pediatric mask, 20 medication cups, 5 replacement filters, mouthpiece, AC adapter, optional rechargeable battery, and carrying case.
You have the option to purchase this ultrasonic nebulizer with or without a rechargeable battery.
Do you find this question helpful?
To prevent possible risk of infection from contaminated medication, clean the nebulizer as follows after each treatment.
To clean: Ensure the plug is disconnected from the device's power socket. Remove the top cover, nebulization chamber, airflow chamber, and medication from the devise. Remove the filter from the cover. Completely empty the conduction liquid from the reservoir. Wipe the main unit with a clean, soft, moistened cloth. Wash the top cover, nebulization chamber, air-flow chamber, mouthpiece, masks, and medication cup in warm, soapy water, and rinse thoroughly. Allow to air dry. Store in a clean, dry location.
Once daily, after washing as above, soak the washed components in a fresh solution consisting of one part warm water and one part white vinegar for 30 minutes followed by a warm water rinse. Allow to air dry. Store in a clean, dry location.
No complaints..Definetly helped in my time of need.